Cookies are not enabled in your browser and may affect website resources. We recommend that you turn it on.
There are several care and restrictions that a blood bank needs to have to give reliability to its patients. After all, have you already stopped to think about how and where the samplers, blood bags of donors, plasma, and others blood products are stored?
These products are stored in refrigerated chambers, thermal chambers, freezers and ultra-freezers, devices especially manufactured to maintain temperature sensitive products. However, how to ensure that the device responsible for storage is working properly and that the device will not be damaged by temperature changes?
It is possible to certify the accuracy of the device by thermal qualification of performance and calibration of temperature meters.
Imagine that a refrigerated chamber that stores blood bags (which must to be stored at temperatures of 2 °C to 6 °C) is measuring with a 4 °C mistake, that the chamber does not return to its original range after a door opening, or that some part of the chamber may be cooling more or less than the determined temperature range. In any of these cases, the product will suffer irreversible damage. This is the importance of making thermal qualifications of performance and calibrations on all equipment that, in general, may interfere with the quality of the final product.
The calibration of temperature sensors are performed by direct comparison with a calibrated temperature standard, where all sensors are introduced in a thermal medium (thermostatic bath or dry block calibrator). The calibration is performed at the desired temperature points. This process creates a calibration certificate. This calibration certificate brings identification, manufactured, serial number (if it has a serial number), sensor type, among others requirements, besides the error and uncertainty that the equipment has presented during a calibration.
Thermal performance qualification in equipment that stores products such as blood, plasma, samples, etc. is performed as follows:
Twelve temperature sensors are installed inside the equipment. One sensor must be close of the sensor of the equipment and the other sensors must be distributed in order to cover all the volume, as we can see in figure below.
The equipment must be qualified with load, that is, it must contain products that are stored daily. Thermal qualification of performance is carried out over 24 hours, during which time the equipment must undergo door openings in order to verify the performance for temperature limits.
It is performed a deliberate shutdown of the energy for five minutes at the end of the qualification to verify if the temperature will exceed the predefined limits in this period.
After the process is finished, the temperature data will be downloaded to generate a certificate with the results. Besides the results, the certificate must bring equipment identification, manufacturer, serial number, model, dimension, graph of temperature readings, photos of the equipment, figure with the distribution of the sensors, conclusion, etc.
NOVUS Metrology Lab uses high-performance, accurate data loggers and Pt100 sensors to measure temperatures during the calibration-qualification process, delivering reliable results.
Periodic equipment calibrations and thermal qualification of performance are important to ensure perfect operation of equipment that stores sensitive products. A reliable company that always uses the same high accuracy equipment offers more reliability to the methods and, therefore, greater safety for health institutions and patients that use blood components and blood products.
96% of all financial transactions are made electronically whose records are stored in large data centers. The circulation of people in these places must be kept to a minimum as they contain sensitive information for the safety of customers. Since 2014, a Brazilian bank has been using NOVUS products with this purpose, putting together a solution consisting of an AirGate Modbus, a RHT-Air and a FieldLogger, with excellent results.
N1050 has a LCD with larger digits, a clear status indication and the lowest power consumption on the market.
It is very difficult to find processes that are not being monitored electronically nowadays. Whether to avoid human error, to meet regulatory requirements or even to reduce costs, the vast majority of companies have already dropped outdated clipboard notes and adopted data logging and monitoring equipment. In order to help choosing the most appropriate device to each process, this article discuss the characteristics of different electronic equipment for monitoring.
Fieldlogger was the chosen solution to measure temperature and vibration of lift pumps at several wastewater treatment plants in São Paulo state. With proper monitoring, it is possible to prevent problems and save on engine maintenance.
TxIsoRail-HRT and TxIsoBlock-HRT bring the robustness of NOVUS devices combined with HART® protocol certification, recognized worldwide as an industry communication standard for smart field devices.
NOVUS is committed to protect the privacy and security of your personal information. The data that you share with NOVUS allows us to provide a better experience when you use our products and services. We guarantee total privacy to your data, used exclusively for communications between NOVUS and you.
In this statement we affirm: